Environmental conditions: Temperature, operating: 10°C to 40°C Temperature, storage: -10°C to 50°C Relative humidity: 25% to 95% (no condensation)
Standards: Safety: IEC (EN) 60601-1, IEC (EN) 60601-1-2(EMC) Performance: IEC (EN) 60601-2-25
Software: EN62304 Protection class: I according to IEC 60601-1(with internal power supply) Applied part: CF according to IEC 60601-1
USB HOST (FS/HS Speed -5Vdc-MAX. 500 mA)
USB DEVICE (FS/HS Speed -5Vdc-MAX. 500 mA)
LAN (RJ-45) (10/100 M/B – UDP/TCP/IP)
USB Host: for upgrading firmware by qualified operator.
USB Device: for transferring data to PC or any host device.
LAN (RJ-45): This port is used in order to communicate with central server or hub for data transfer.
One can use this port for communicating with hospital PACS system.
PACS: Picture Archiving and Communication System
WiFi: is used in the place which wiring is not possible. This method is used for transferring data from device to sever of reference PC.
One can use this method for communicating with hospital PACS system.
SD Card: for saving recorded ECG signal data on SD card.